The GAPS time-of-flight system (ToF) provides measurements of a particle's velocity, charge, and trajectory. The ToF consists of numerous 1.6 or 1.8-meter-long plastic scintillator paddles assembled into two components, an outer "umbrella" section and an inner "cube" section, separated by ~1 meter. The cube surrounds the Si(Li) tracker. Paddles are wrapped to be light-tight, and are held in place with a carbon fiber structure. Silicon photomultipliers (SiPMs) are coupled to both ends of each paddle, and are connected to custom preamplifier and readout boards. The readout boards incorporate the Paul Scherrer Institute DRS4 chip, allowing for sample rates of several GHz. A computer organizes and compresses the data from the readout boards for both local storage and telemetry.
An incoming (or outgoing) charged particle produces photons that undergo total internal reflection within the scintillator and are observed by the SiPMs at the paddle ends. Photon arrival times at each paddle end can be determined with sub-nanosecond accuracy. The relative timing between paddle ends gives us the position of a particle hit, and hits on different paddles determine the particle's trajectory. Velocity can then be derived using the timing difference between paddle hits. Charge is estimated using the energy deposited in each paddle, the velocity, and the trajectory.
The ToF also triggers readout of the whole instrument, at a rate of a few 100 Hz. The trigger decision algorithm is designed to initiate readout when particles of interest (e.g. antiprotons, antideuterons) enter the instrument, while simultaneously vetoing background particles (e.g. protons, helium nuclei, carbon nuclei). The trigger decision is made by a master trigger board, which combines information from 26 local trigger boards, and incorporates a particle's speed, charge, and the total number of ToF paddles hit. In general, particles of interest will have lower speed, lower charge, and more ToF paddle hits than background particles. The trigger decision must be made within ~300 nanoseconds.